Email required Address never made public. How is switching achieved between them? I am a amateur radio fan and just bought a used Hantek DSO However, like Fabio, be assured that I am very appreciative for all your efforts! The lead is easy to access the existing automotive wiring harness interface, convenient to read all kinds of sensor signal. Especially when it got warm, it gave me the red blinking led.
|Date Added:||20 January 2010|
|File Size:||49.20 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I described the problem in more detail here: Or is the needed to pull that trick? The triggering is rock solid.
Hantek DSO Oscilloscope | eBay
Could you estimate how often do you see the wrong waveform when sampling the 1kHz test signal? I agree, by the way, that what the sigrok folks have written is a lot easier to read than the low level openhantek code. Happy to hear it worked out!
I forgot that I set the averages to 2! Second — the TriggerCounter is activated and counts to zero.
New Hantek 40MHz PC Based USB Oscilloscope – DSO
When I switched to the original openhantek code base from sourceforge, everything worked as it should. I have verry good soldering skills, but have no real idea about ksb inner workings of the scope, so thanks for the useful info. What distro of Linux are you using?
The weird thing is that the glitch at MHz appeared to be software related somehow: I am on the verge of giving up! That might be nice info to add to your main entry.
When I move the trigger slightly to the left the jitter is much diminished. A trigger event occurs and two things happen. Thanks for the quick reply — this seems like a good general purpose piece of kit. Good idea putting the problem on a video! High-Voltage x Probe T Anyway, I have now done some updates and bugfixes to the software, branched from the Dec 4 version. Having said that, there is no reason that I can think of why Hantek would want to do this — after all, they sell hardware and give away software.
I wonder if you can help me not to do so.
Multimeter Banana Socket HT From the high res picture on sigrok website all of those seems to be SST http: The lowest-priced brand-new, unused, unopened, undamaged item in its original packaging where packaging is applicable. Show More Show Less. I guess I just landed on a buggy commit last time.
The case has a grill on the bottom under the CPLD to allow air flow but nothing on the top. These are crazy to my eyes. Any advise on which competent could be faulty. The triggers are tricky too. I have got DSO Perhaps the confusion originated when they added support for the A which has a k deep buffer of relatively slow memory.
Thanks for all the useful information and good conversation! You are commenting using your Twitter account. The ADCs are humming along filling and refilling the 32K of ram one for each channel in this example.